I was wondering if anyone else had this problem or if there's some trick to it. The time now is 02:56 PM. Your cache administrator is webmaster. I'd get error 5 also and I know there was enough blood on the strip. Source
My problem is more the lancet device. This site was last updated on December 01, 2016. This site is published by LifeScan, Inc, which is soley responsible for its content. All trademarks are the property of their respective owners. Debbie Type 2 since 1995 Started Insulin December 2003 Cozmore pump Started 3/22/07 using Novolog Animas Ping - started 06/09/2009 MM Revel - Started 4/22/10 MOD Squad Reply With Quote 4/09/07
AdvaMed Guidelines. I tried using it and out of about 15 tries I got one good reading in the morning, but when I tried after lunch, it just wouldn't work. If this message continues to appear, contact OneTouch Customer Service at 1 888-567-3003.If you applied the sample incorrectly, review the blood application process or control solution testing and repeat the test One Touch Ultra Mini Error Code C25 All rights reserved.
I take care to lance and get a sufficient drop and quickly position it to fill the channel all at once. One Touch Ultra Mini How To Use Your cache administrator is webmaster. I've noticed if it actually does the test, it will give me a lower reading. Last Jump to page: Thread Tools Email this Page… 4/09/07 #1 Krisb0701 Retired T1 Pump & CGM & Celiac Age 31 Join Date March 2007 Location Boston, MA Posts > 100
It is intended for visitors from the United States. One Touch Select Simple Manual I've used many lifescan meters and prefer them above all else. The only thing worse is when my pump site is not absorbing the insulin I bolus after I discover my suger is already high and it goes higher while I feel Get about as much blood as a circle the size of the bottom half of an 8 in a serial number on US currency.
Life scan keeps telling me that I must just be getting a bad batch. https://www.diabetesdaily.com/forum/testing-blood-sugar/3472-does-anyone-have-issues-their-one-touch-meter/ They have a terrible time finding a vein (?) and many times it take the multiple tries to get blood. One Touch Ultra Error 2 Review the blood application process or control solution testingThe meter battery is low but there is enough power to perform a test.When the flashing battery icon first appears, there is enough One Touch Select Simple Error Codes BTW the doctor had given me 3 containers of strips, and I had tried strips from all 3 and gotten nothing but errors.
You need to make sure you lance and position the blood drop to the side of the strip as shown in the manual on page 38. http://vbview.net/one-touch/one-touch-error-4.php I got it in July and didn't get around to using it at first, but when I did (a few weeks ago), I kept getting an error message, #4, something about Instead of making 2 types of strips, they stuck with this one where either way both types of problems will lose out and get that damned error 5 reading. orThere may be a problem with the meter.If you tested in a cool environment, repeat the test in a warmer environment with a new test strip. One Touch Select Simple Error 8
People with cholestoral problems most likely caused by treating their bodies like garbage dumps take blood thinning medications. So anyway, the company sent me a new monitor, but no new strips. I hope you manage to work things out with yours Kris. have a peek here So I phoned the company, and they had me try some things over the phone.
It is extremely annoying and expensive and I am quite surprised that such a faulty device was allowed on the market. One Touch Verio Error 4 So anyway, the company sent me a new monitor, but no new strips. Another suggestion in case there may be a problem with the way you are doing the test.
Featuring Advanced Glide® Control System. only error i get is when i dont pay attention and get the blood on too soon, or dont fill the strip. The error #4 is a strip fill error. One Touch Ultra Mini Battery A Freestyle monitor comes with the Cozmo you are getting.
Possible causes are test strip damage or an incompletely filled out confirmation window.Repeat the test with a new test strip. Just a thought. Discard the test strip and re-start the test process.Step 3 Wait for the confirmation window to fill completelyThe blood drop will be drawn into the narrow channel and the confirmation window http://vbview.net/one-touch/one-touch-error-3.php By Marvin Gardens in forum Type 2 Diabetes Replies: 5 Last Post: 3/18/07, 03:33 PM Keeping in Touch By Rascalette in forum Type 2 Diabetes Replies: 2 Last Post: 12/23/06, 05:46
Get additional information. © 2005- 2016 Diabetes Daily, LLC. Working together to help you perform at your best.Learn MoreBuy NowOneTouch Ultra® 2OneTouch Ultra® 2A fast and simple way to see the effect of food on your blood sugar results.Learn MoreBuy Get additional information. © 2005- 2016 Diabetes Daily, LLC. Now I just use a screen protection sheet.
Hopefully they will if they cover that meter. Improper use may cause an inaccurate result without producing a warning message.MessageWhat it meansWhat to doYou may have a very low blood sugar level (severe hypoglycemia), lower than 20 mg/dL.This may It worked the first time, but no longer an excuse. That being said, I found the Verio Gold strips more sensitive to the positioning and timing of getting the blood on the strip.
Or call: 1-844-807-8936MEDICARE PART BWe've got you covered.Find approved suppliers. I am fed up. I keep wanting the One Touch to work, but it really is awful. If you accidently touch your finger, or have too little or too much blood.
Originally Posted by slanoue I was wondering if anyone else uses the OneTouch Verio IQ, and if so, whether you've had problems with it? It usually takes three to four strips before I get a result. I get my share of Error 5's. Suzanne Lanoue [/url] Reply With Quote 9/09/14 #2 bsc D.D.
Contact Us Legal Privacy Sitemap LifeScanThis information is for general background purposes and is not a substitute for medical advice or treatment for specific conditions.